Using gas chromatography, the model airTOXIC BTX PID measures the amount of Benzene, Toluene, Ethylbenzene and Xylenes in the air from sub-ppb levels up to 1,000 ppb.
Teledyne API 100 UV Fluorescent SO2 Analyser
Teledyne API 100A UV Fluorescent SO2 Analyser
Teledyne API 100E UV Fluorescent SO2 Analyser
Teledyne API 101 UV Fluorescent SO2 Analyser
Teledyne API 101A UV Fluorescent SO2 Analyser
Teledyne API 200 chemiluminescent NOx analyser
Teledyne API 200A chemiluminescent NOx analyser
Teledyne API 200AU chemiluminescent NOx analyser
Teledyne API 200E chemiluminescent NOx analyser
Teledyne API 200EU chemiluminescent NOx analyser
Teledyne API 200EU chemiluminescent NOx analyser - blue light
Teledyne API 200UP chemiluminescent NOx analyser
Teledyne API 300 gas filter correlation CO analyser
Teledyne API 300A gas filter correlation CO analyser
Teledyne API 300E gas filter correlation CO analyser
Teledyne API 400 UV photometric O3 analyser
Teledyne API 400A UV photometric O3 analyser
Teledyne API 400E UV photometric O3 analyser
Teledyne API M100E UV Fluorescent SO2 Analyser
Teledyne API M200E chemiluminescent NOx analyser
Teledyne API M300E gas filter correlation CO analyser
Teledyne API M400E UV photometric O3 analyser
Teledyne API T100 UV Fluorescent SO2 Analyser
Teledyne API T200 chemiluminescent NOx analyser
Teledyne API T200U chemiluminescent NOx analyser
The Model T200U Trace-level NO/NO2/NOx analyzer has been developed specifically to address the challenges of low level monitoring as required, for example, in the US NCore network. It uses the proven chemiluminescence principle and is designed to allow ultra-sensitive measurements with a lower detectable limit of 50 ppt while still meeting the requirements for use as a US EPA compliance analyzer. The Model T200U combines a gold plated reaction cell with an enhanced performance pump, special low noise photomultiplier tube, and unique signal conditioning to provide exceptional sensitivity. In addition, a pre-reactor separates the NO-O3 reaction from background chemiluminescence to allow accurate auto-zero of the analyzer. All T Series instruments offer an advanced color display, capacitive touch screen, intuitive user interface, flexible I/O, and built-in data acquisition capability. All instrument set up, control and access to stored data and diagnostic information is available through the front panel, or via RS232, Ethernet, or USB com ports, either locally or by remote connection. The Model T200U comes NumaView™ Software. NumaView™ Remote PC Software allows for a remote connection with virtual interface and data downloading capability to analyzers operating NumaView™ Software.
Teledyne API T300 gas filter correlation CO analyser
Teledyne API T400 UV photometric O3 analyser
Environnement S.A. Model 1003-RS under DASIBI license
SINTECH SPECTRAS BTX GC 855 series undetermined
SYNTECH SPECTRAS GC 955 series undetermined
GRIMM model EDM 107 for PM10, PM2.5, PM1 and TC
The GRIMM model 107 monitor is designed to measure particle size distribution and particulate mass based on a light scattering measurement of individual particles in the sampled air
GRIMM model EDM 180-C for PM10 and PM2.5
GRIMM model EDM 180-D for PM10, PM2.5, PM1 and TC
GRIMM model EDM 180-MC for PM10, PM2.5, PM1, TC and 31 size channels
GRIMM model EDM 365 for PM10, PM2.5, PM1 and TC
IL-Punkkinen for NO2/NO hourly measurements
Device based on the modified automated Saltzman method by Finnish Meteorological Institute.
TEOM 1405DF with FDMS Dichotomous monitor for PM2.5 & PM10
Agilent HP 6890 (used for benzene and toluene measurement)
System for analysis of the Ozone precursor VOC
The airmoVOC BTX is an automatic gas chromatograph dedicated to the analysis and monitoring of trace amounts of BTEX compounds (Benzene, Toluene, Ethylbenzene, m-, p-, and o-Xylenes),
The airpointer‘s modular design comprising a base unit, analyzing modules and sensor modules allows for a configuration according to different application requirements.
Andersen/GMW Model FH621-N Beta Monitor
Andersen Instruments Model FH62I-N PM10 Beta Attenuation Monitor consisting of the following components: FH101 Vacuum Pump Assembly; FH102 Accessory Kit; FH107 Roof Flange Kit; FH125 Zero and Span PM 10 Mass Foil Calibration Kit; FH62I Beta Attenuation 19-inch Control Module; SA246b PM10Inlet (16.7 liter/min) or the louvered inlet specified in 40 CFR 50 Appendix L, Figs. L-2 thru L-19; operated for 24-hour average measurements, with an observing time of 60 minutes, the calibration factor set to 2400, a glass fiber filter tape, an automatic filter advance after each 24-hour sample period, and with or without either of the following options: FH0P1 Indoor Cabinet; FH0P2 Outdoor Shelter Assembly
Andersen/GMW generic High-Volume Air Sampler
Bendix/Combustion Engineering Model 8101-C Oxides of Nitrogen Analyze
Bendix or Combustion Engineering Model 8101-C Oxides of Nitrogen Analyzer operated on a 0-0.5 ppm range with a Teflon sample filter (Bendix P/N 007163) installed on the sample inlet line.
Columbia Scientific Industries Models 1600
CSI Model 1600 Oxides of Nitrogen Analyzer operated on a 0-0.5 ppm range with a Teflon sample filter (CSI P/N M951-8023)installed on the sample inlet line
Thermo Andersen Series FH 62 C14 Continuous PM10 Ambient Particulate Monitor
Thermo Andersen Series FH 62 C14 Continuous PM10 Ambient Particulate Monitor and Thermo Scientific Model 5014i Beta (5014i Beta), Continuous Ambient Particulate Monitor,” operated for 24-hour average measurements, with the specified 10-micron EPA PM10inlet (or alternative louvered PM10inlet meeting 40 CFR 50 Appendix L specifications), inlet connector, sample tube with heater, roof flange kit, mass foil kit, pump kit, sample filter tape; with operational settings of 1000 L/h (16.67 L/min) sample flowrate, daily filter change, auto filter change at volumetric flow <950 L/h, auto filter change at mass >1500 micrograms, and factory default calculation mode settings operated with software version 1.07. Operated, calibrated and serviced according to the appropriate Operator Manual
Columbia Scientific Industries Models 5600
CSI Model 5600 Oxides of Nitrogen Analyzer operated on a 0-0.5 ppm range, with any signal integration time in the range of 20 to 99 seconds, with a Teflon sample filter (CSI P/N M951-8023) installed on the sample inlet line
Environnement S.A. Model AC21M NO2 Analyzer
Environnement S.A. Model AC30M NO2 Analyzer
Environnement S.A. Model AC31M NO2 Analyzer
Environnement S. A. Model AC31M Chemiluminescent Nitrogen Oxide Analyzer operated with a full scale range of 0 -500ppb, at any temperature in the range of 15°C to 35°C, with a 5-micron PTFE sample particulate filter, with the following software settings: Automatic response time ON; Minimum response time set to 60 seconds (RT 2); and with or without any of the following options: Internal Permeation Oven; Connection for Silica Gel Dryer; RS232-422 interface; EV3 valve; Internal Printer
Environnement S.A. Model AC32M NO2 Analyzer
Environnement S. A. Model AC32M Chemiluminescent Nitrogen Oxides Analyzer operated with a full scale range of 0 -500 ppb, at any temperature in the range of 10°C to 35°C, with a 5-micron PTFE sample particulate filter, with response time setting 11 (automatic response time), and with or without the following option:Internal permeation oven.
Environnement S.A. Model AC32E NO2 Analyzer
Environnement S.A. Model AF20M SO2 Analyzer
Environnement S.A. Model AF21M SO2 Analyzer
Environnement S.A Model AF21M Sulfur Dioxide Analyzer operated on a range of 0-0.5 ppm with a response time coefficient setting of 01, a Teflon ilter installed in the rear - panel filter assembly, and with or without any of the following options: Rack Mount/Slides; RS-232-C Interface
Environnement S.A. Model AF22M SO2 Analyzer
Environnement S.A Model AF22M UV Fluorescence Sulfur Dioxide Analyzer operated with a full scale range of 0 - 500 ppb, at any temperature in the range of 10°C to 35°C, with a 5-micron PTFE sample particulate filter, with a response time setting of 11 (Automatic response time), with the automatic “ZERO-REF” cycle ON and set for activation every 24 hours, and with or without either of the follow ing options: Permeation oven, Rack mount slides
Environnement S.A. Model AS32M Nitrogen Dioxide Analyzer
Environnement S.A. Model CO10M CO Analyzer
Environnement S.A. Model CO11M CO Analyzer
Environnement S.A. Model CO11M Ambient Carbon Monoxide Analyzer operated on a full scale range of 0 - 50 ppm, at any temperature in the range of 15 °C to 35°C, with a 5-micron PTFE sample particulate filter, with the following software settings:Automatic response time ON; Minimum response time set to 40 seconds (RT 13); Automatic ZERO-REF cycle programmed every 24 hours; and with or without any of the following options: RS232-422 Serial Interface; Internal Printer
Environnement S.A. Model CO12E CO Analyzer
Environnement S.A. Model CO12M CO Analyzer
Environnement S.A Model CO12M Gas Filter Correlation Carbon Monoxide Analyzer operated with a full scale range of 0 - 50 ppm, at any temperature in the range of 10°C to 35°C, with a 5-micron PTFE sample particulate filter, with response time ON, and with the automatic “ZERO-REF” cycle either ON or OFF
Environnement S.A. Model MP101M (may be used for PM10 and PM2.5)
Environnement S. A. Model MP101M with PM10 or PM2.5 Beta Gauge Monitor,” configured with the louvered PM 10 or PM2.5 inlet specified in 40 CFR 50 Appendix L or its flat-topped predecessor version and one of the three optional temperature-regulated sampling tubes (RST), and operated with a full scale measurement range of 0 - 0.500 mg/m3 (0 - 500 g/m3), with the sample flow rate set to 1.00 m3/h and flow regulation set to yes , the “norms selection” set to m3(actual volume), the “cycle” set to 24 hours, the “period” set to none, and the “counting time” set to 200 seconds
Environnement S.A. Model MP2.5M undefined
Environnement S.A. Model O341M UV Ozone Analyzer
Environnement S.A. Model O341M UV Photometric Ozone Analyzer operated on a full scale range of 0 - 500 ppb, at any temperature in the range of 15°C to 35°C, with the response time set to 50 seconds, and with or without any of the following options: Internal Ozone Generator; Span External Control; RS232 - 422 Serial Interface; Internal Printer.
Environnement S.A. Model O342E UV Ozone Analyzer
Environnement S.A. Model O342M UV Ozone Analyze
Environnement S.A Model O342M UV Photometric Ozone Analyzer,” operated with a full scale range of 0 - 500 ppb, at any temperature in the range of 10°C to 35°C, with a 5-micron PTFE sample particulate filter, with response time setting of 11 (Automatic response time), and with or without any of the following options: c) Internal ozone generator, d) Span external control (zero/span solenoid valve).
Environnement S.A. SANOA Multigas Longpath Monitoring System
Environnement S.A. Model SANOA Multigas Longpath Air Quality Monitoring System consisting of a receiver, one or more projectors, interface unit, a user-provided control unit computer running the SANOA VisionAIR software, and associated incidental equipment; configured for measuring SO2, with the temperature control and internal calibration cell options installed, operated with a measurement range of 0 to 0.5 ppm, over an installed monitoring path length of between 27 and 500 meters, within an ambient air temperature range of -30 to +45°C,with a measurement (integrating) time of 180 seconds, and with or without external temperature and barometric pressure sensors or any of the following options: external (meteo) input connection, series 1M bus connection, OGR type projector, analog outputs.
Horiba model APHA 360E hydrocarbons analyser
Horiba model APHA 360 hydrocarbons analyser
Horiba model APHA 370 hydrocarbons analyser
MCV CAV-A/MSb Captador d'Alt Volum Seqüencial
Meloy Model SA 700 Fluorescence Sulfur Dioxide Analyze
Meloy Model SA 700 Fluorescence Sulfur Dioxide Analyzeroperated on the 0-250 ppb1, the 0-500 ppb, or the 0-1000 ppb range with a time constant switch position of either 2 or 3.The analyzer may be operated at temperatures between 20°C and 30°C and at line voltages between 105 and 130 volts, with or without any of the following options: FS-1 Current Output; FS-2 Rack Mount Conversion; FS-2A Rack Mount Conversion; FS-2B Rack Mount Conversion; FS-3 Front Panel Mounted Digital Meter; FS-5 Auto/Manual Zero/Span With Status; FS-6 Remote/Manual Zero/Span With Status; FS-7 Auto Zero Adjust.
Multi-gas Micro Monitoring station (MMS) Portable version
Opsis AR500 Open path monitor with ER 110
Opsis AR500 Open path monitor with ER 120
Opsis AR500 Open path monitor with ER 150
Opsis SM200 Particulate Analyser with PM2.5 & PM10 Heads
Osiris air monitoring device measures airborne particles (TSP, PM10, PM2.5 and PM1). MCERTS certified for PM10
PHILIPS K50109/00 Gas Filter Correlation CO analyser
PHILIPS K50110/00 UV Photometric O3 analyser
PHILIPS K50206/00 Pulsed Fluorescence SO2 analyser
Ecotech serinus 40 used for NOx-NO2-NO modified to measure NOH3 (including a converter)
EC/OC measurements Sunset Laboratory Inc Thermal / Optical Carbon Analyser (near-real time)
EC/OC measurements Sunset Laboratory Inc Thermal / Optical Carbon Analyser - operated near real time
Tekran mercury analyser 2537A. Elemental mercury is sampled by placing the 1130 before the main 2537A instrument. The respective constituents are sampled from the air using either the 1130 add-on for reactive gaseous mercury or the 1135 add-on for particulate bound mercury.
Thermo model 14B chemiluminescence NO-NO2-Nox
Thermo model 14B/E chemiluminescence NO-NO2-Nox
Thermo model 42i NO/NOx analyser - blue light
Thermo model 43h pulsed fluorescence SO2 analyser
Thermo model 43i-TLE (Enhanced Trace Level)
Thermo model 48 CO analyser (gas filter correlation)
Thermo model 48i-TLE (Enhanced Trace Level)
This is a reference photomer, not an analyser
Thermo 5014i Beta Continuous Ambient Particulate Monitor
Measure PM-10, PM-2.5 or PM-1 mass concentrations with the Thermo Scientific™ 5014i Beta Continuous Ambient Particulate Monitor. The 5014i distinguishes itself from other beta measurement methods by utilizing a continuous (non-step wise) mass measurement with a proven industry standard which provides for long-term unattended operation. To accurately address potential water bias and volatile loss, the Dynamic Heating System allows the user to hold the sample temperature at a fixed value or below a relative humidity threshold.
Thermo model 5030 SHARP Particular Monitor with PM10 & PM2.5 heads
The Thermo Scientific™ 5030 SHARP continuously measures the mass concentration of particulate utilizing a combination of beta attenuation and light scattering technology. The only continuous (non-step wise) hybrid ambient monitor that provides high resolution measurement of airborne particulates with excellent short and long term precision.